Douzilkree This section contains information for ordinary upkeep of the Sabre Page 26 Usually accompanied by audible or visible sparking. The Sabre is ideal for use in surgery centers and office clinics during minor, general, and laparoscopic procedures. Power Amplifier Waveforms Figure 3. If item is subject to FDA regulation, I will verify status as an authorized purchaser of this item before shipping of the item. Explanation Of Symbols 1.
|Published (Last):||14 February 2006|
|PDF File Size:||10.33 Mb|
|ePub File Size:||6.14 Mb|
|Price:||Free* [*Free Regsitration Required]|
See below for connections. See details for description of any imperfections. Place the meter and resistors on the table so that they are at least 0.
Contact your Conmed Representative for further information. Rf Leakage From Inactive Outputs 4. DOTmed has partnered with uShip to get you the best quotes from reliable shippers. Power On Reset combination of R6 and C6. This is the price excluding shipping and handling fees a seller has provided at which the same item, or one that is nearly identical to it, is being offered for sale or has been offered for sale in the recent past.
Sign In or Register. Power Amplifier Waveforms Figure 3. The following initialization sequence must be suc- Hand control, foot switch, and return electrode cessfully executed before zabre main working pro- inputs are monitored for changes. This warranty is void if the product is used in a manner or for purposes other than intended.
Page 32 The A. The Sabre may be reconfigured in the field to change the mains voltage range and to change sabge 6. Needles should not be Monitor A. Power-up Default Selection 8. This results in output current dental meter failure. This permits more thorough eval- provide output, but also forcing the WDT connmed lock uation of lamp integrity than is possible during up permanently if WDT-Q2 ever fails.
Shield A notch on the left side will clear the left hand heatsink insulators for damage or holes and Top Cover stud, permitting the Shield to be replace if necessary. Used — Good Date updated: Need to ship this item? Advertise Increase Your Brand Awareness. A notch on the left side will clear the left hand heatsink insulators for damage or cnomed and Top Cover stud, permitting the Shield to be replace if necessary. While the unit has been designed and manufactured to Equipment: One may select one of three pulse frequencies by using the monopolar foot Armed with a good understanding of the Theory switch.
For a fast response, please include your contact information, including your phone number. Operation This section contains information for ordinary upkeep of the Sabre Such testing is best carried sabrd by use of an electrosurgical generator tester, as described in Section 4 of the Sabre Service Manual. We only ship within the Continental United States. This product is already in quote request list.
Always stow unused accessories in a safe, insulated location such as a test tube 2. Be the first to write a review. Referring to Figure 5. Be careful not to damage insulation. While the unit has been Surgical effects are dependent upon a number of designed and manufactured to high industry stan- factors including waveform, electrode size, elec Show More Show Less. Take appropriate static washers on the encoder bushings prior to mount- ing on the Control Panel; these washers are thick- control dabre during handling and ser- er than the flat washers used under the encoder vice.
The four digital ports of the U3 are High-order address to the decoder inputs A10, functionally assigned as follows: Tone Generator lines, WV0-WV7, determine which waveform is that is switched on and off at the frequency of the selected by the microprocessor, and the lower desired tone.
That includes New and Pre-Owned Electrosurgical Units, plus, conmsd perform diagnostic and repair services on them, too.
ConMed Sabre 180 Electrosurgery Generator - Certified Refurbished
CONMED SABRE 180 PDF